The objective of this study was to perform simulated-use testing as well as a clinical study to assess the efficacy of the EVOTECH® Endoscope Cleaner and Reprocessor (ECR) cleaning for flexible colonoscopes, duodenoscopes, gastroscopes and bronchoscopes. The main aim was to determine if the cleaning achieved using the ECR was at least equivalent to that achieved using optimal manual cleaning.
Simulated-use testing consisted of inoculating all scope channels and two surface sites with Artificial Test Soil (ATS) containing 108 cfu/mL of Enterococcus faecalis, Pseudomonas aeruginosa and Candida albicans. Duodenoscopes, colonoscopes, and bronchoscopes (all Olympus endoscopes) were included in the simulated use testing. Each endoscope type was tested in triplicate and all channels and two surface sites were sampled for each scope. The clinical study evaluated patient-used duodenoscopes, bronchoscopes, colonoscopes, and gastroscopes (scopes used for emergency procedures were excluded) that had only a bedside flush prior to being processed in the ECR (i.e. no manual cleaning). There were 10 to 15 endoscopes evaluated post-cleaning and to ensure the entire ECR cycle was effective, 5 endoscopes were evaluated post-cleaning and post-high level disinfection. All channels and two external surface locations were sampled to evaluate the residual organic and microbial load. Effective cleaning of endoscope surfaces and channels was deemed to have been achieved if there was < 6.4 μg/cm2 of residual protein, < 1.8 μg/cm2 of residual hemoglobin and < 4 Log10 viable bacteria/cm2. Published data indicate that routine manual cleaning can achieve these endpoints so the ECR cleaning efficacy must meet or exceed these to establish that the ECR cleaning cycle could replace manual cleaning
In the clinical study 75 patient-used scopes were evaluated post cleaning and 98.8% of surfaces and 99.7% of lumens met or surpassed the cleaning endpoints set for protein, hemoglobin and bioburden residuals. In the simulated-use study 100% of the Olympus colonoscopes, duodenoscopes and bronchoscopes evaluated met or surpassed the cleaning endpoints set for protein, and bioburden residuals (hemoglobin was not evaluated).
The ECR cleaning cycle provides an effective automated approach that ensures surfaces and channels of flexible endoscopes are adequately cleaned after having only a bedside flush but no manual cleaning. It is crucial to note that endoscopes used for emergency procedures or where reprocessing is delayed for more than one hour MUST still be manually cleaned prior to placing them in the ECR.