One of the major uses of cone beam computed tomography (CBCT) is presurgical implant planning. Linear measurement is used for the determination of the quantity of alveolar bone (height and width). Linear measurements are used in orthodontic analysis and definition of jaw tumor size. The objective of this study is to evaluate the accuracy of the linear measurement of CBCT (Newtom VG) in the axial and coronal planes, with two different voxel sizes.
Materials and Methods:
In this accuracy diagnostic study, 22 anatomic landmarks in four dry human skulls were marked by gutta-percha. Fifteen linear measurements were obtained using a digital caliper. These were considered to be the gold standard (real measurement). The skulls were scanned by CBCT (Newtom VG) at two settings: (a) Voxel size 0.3 mm (b) voxel size 0.15 mm High Resolution (HR). The radiographic distance measurements were made in the axial and coronal sections by three observers. The radiographic measurements were repeated two weeks later for evaluation of intraobserver reliability. SPSS software version 17 was used for data analysis. The level of significance was considered to be 5% (P ≤ 0.05).
The mean differences of real and radiographic measurements were -0.10±0.99 mm in the axial sections, -0.27±1.07 mm in the coronal sections, +0.14±1.44 mm in the axial (HR) sections, and 0.02±1.4 mm in the coronal (HR) sections. The intraclass correlation (ICC) for CBCT measurements in the axial sections was 0.9944, coronal sections 0.9941, axial (HR) sections 0.9935, and coronal (HR) sections 0.9937. The statistical analysis showed high interobserver and intraobserver reliability (P ≤ 0.05).
CBCT (Newtom VG) is highly accurate and reproducible in linear measurements in the axial and coronal image planes and in different areas of the maxillofacial region. According to the findings of the present study, a CBCT scan with a larger voxel size (0.3 mm in comparison to 0.15 mm) is recommended when the purpose of the CBCT scan is to measure linear distances. This will result in lower patient radiation dose and faster scan time.
Cone-beam computed tomography; implant; measurement; skull
The objective of this study was to determine the ability of two flat panel cone beam CT (CBCT) devices to identify demineralized bone and bone transplants in vivo and in vitro.
Datasets from patients with autologous bone grafts (n = 9, KaVo 3DeXam (KaVo, Biberach, Germany); n = 38, Accuitomo 40 (Morita, Osaka, Japan)) were retrospectively evaluated. Demineralized and non-demineralized porcine cancellous bone blocks were examined with the two CBCT devices. A SawBone® skull (Pacific Research Laboratories, Vashon, WA) was used as a positioning tool for the bone blocks. Descriptive evaluation and image quality assessment were conducted on the KaVo 3DeXam data (voxel size 0.3 mm) using the OsiriX viewer as well as on the Morita Accuitomo data (voxel size 0.25 mm) using proprietary viewer software.
Both in vivo and in vitro, the descriptive analysis of the images of the two devices showed well-visualized bone transplants with clearly defined cancellous bones and well-defined single bone trabeculae in all cross-sections. In vitro, demineralized samples showed lower radiographic opacity but no significant loss of quality compared with fresh bone (P = 0.070). Single cancellous bone trabeculae were significantly better visualized with the Morita 3D Accuitomo device than with the KaVo 3DeXam device (P = 0.038).
Both the KaVo 3DeXam and Morita 3D Accuitomo devices produce good-quality images of cancellous bones in in vivo remodelling as well as after in vitro demineralization.
cone beam computed tomography; bone augmentation; implantology
The aim of this investigation was to optimize the positioning and size of the scanned field of view (FOV) in cone beam CT (CBCT) scanners using a practical external alignment device fitted in the patient's mouth in order to train radiographers and reduce radiation dose to the patient. This is particularly challenging when using small FOVs to cover small volumes of interest.
Test objects were positioned and scanned using the aligner to show that the design and geometry were correct and help the radiographer to superimpose the scanner and the volume of interest axis of rotation. An in vivo study was then undertaken comparing the accuracy of patient positioning when using the aligner, instead of scouts, to position the patient for small FOV (cylinders of 4–8 cm height and 4–8 cm diameter) dental scans. The scanners used were the Accuitomo F170 CBCT scanner (Morita, Kyoto, Japan) and the iCAT Next Gen CBCT scanner (Imaging Sciences, Hatfield, PA).
There was no significant difference in positioning the patient when using the aligner compared with the scout images.
It is possible to rely on the aligner for patient positioning for a small volume scan and therefore spare the radiation dose associated with scout imaging.
dental; radiation dosage; harm reduction; best practice analysis
We aimed to assess the diagnostic accuracy of different cone beam CTs (CBCTs) and the influence of field of view (FOV) in diagnosing simulated periapical lesions.
6 formalin-fixed lateral mandibular specimens from pigs were used for creating 20 standardized periapical bone defects. 18 roots were selected for the control group. Three CBCT devices [Accuitomo 3D® (Morita, Kyoto, Japan), NewTom 3G (Quantitative Radiology, Verona, Italy) and Scanora® (Soredex, Tuusula, Finland)] and three FOVs (NewTom 3G® FOV 6, 9 and 12 inches) were used to scan the mandible. Five observers assessed the images, using a five-point probability scale for the presence of lesions. Specificity, sensitivity and areas under the receiver operating characteristic (ROC) curves were calculated.
Sensitivity ranged from 72% to 80%. Specificity ranged from 60% to 77%. A difference in scoring between Scanora and the other two devices existed only in the control group. ROC analysis for different FOVs showed a decreased sensitivity with an increasing FOV, but this difference was not significant.
In the control group, there was a difference between the CBCT devices regarding their specificity. FOV size did not show any difference in diagnostic performance. In cases in which conventional radiographic methods in combination with clinical evaluation are not sufficient, CBCT may be the method of choice to assess periapical pathology. CBCT examinations should be complementary to a clinical examination and FOV adaptation can be utilized to keep the dose to the patient as low as possible.
cone beam computed tomography; apical lesion; in vitro; sensitivity; field of view
The purpose of this study was to compare mandibular linear distances measured from cone beam CT (CBCT) images produced by different radiographic parameter settings (peak kilovoltage and milliampere value).
20 cadaver hemimandibles with edentulous ridges posterior to the mental foramen were embedded in clear resin blocks and scanned by a CBCT machine (CB MercuRayTM; Hitachi Medico Technology Corp., Chiba-ken, Japan). The radiographic parameters comprised four peak kilovoltage settings (60 kVp, 80 kVp, 100 kVp and 120 kVp) and two milliampere settings (10 mA and 15 mA). A 102.4 mm field of view was chosen. Each hemimandible was scanned 8 times with 8 different parameter combinations resulting in 160 CBCT data sets. On the cross-sectional images, six linear distances were measured. To assess the intraobserver variation, the 160 data sets were remeasured after 2 weeks. The measurement precision was calculated using Dahlberg's formula. With the same peak kilovoltage, the measurements yielded by different milliampere values were compared using the paired t-test. With the same milliampere value, the measurements yielded by different peak kilovoltage were compared using analysis of variance. A significant difference was considered when p < 0.05.
Measurement precision varied from 0.03 mm to 0.28 mm. No significant differences in the distances were found among the different radiographic parameter combinations.
Based upon the specific machine in the present study, low peak kilovoltage and milliampere value might be used for linear measurements in the posterior mandible.
cone-beam computed tomography; dental implants; mandible
This study was performed to determine the accuracy of linear measurements on three-dimensional (3D) images using multi-detector computed tomography (MDCT) and cone-beam computed tomography (CBCT).
Materials and Methods
MDCT and CBCT were performed using 24 dry skulls. Twenty-one measurements were taken on the dry skulls using digital caliper. Both types of CT data were imported into OnDemand software and identification of landmarks on the 3D surface rendering images and calculation of linear measurements were performed. Reproducibility of the measurements was assessed using repeated measures ANOVA and ICC, and the measurements were statistically compared using a Student t-test.
All assessments under the direct measurement and image-based measurements on the 3D CT surface rendering images using MDCT and CBCT showed no statistically difference under the ICC examination. The measurements showed no differences between the direct measurements of dry skull and the image-based measurements on the 3D CT surface rendering images (P>.05).
Three-dimensional reconstructed surface rendering images using MDCT and CBCT would be appropriate for 3D measurements.
Reproducibility of Results; Three-Dimensional Image; Skull
The aim of this study was to determine the geometric accuracy of cone beam CT (CBCT)-based linear measurements of bone height obtained with the Galileos CBCT (Sirona Dental Systems Inc., Bensheim, Hessen, Germany) in the presence of soft tissues.
Six embalmed cadaver heads were imaged with the Galileos CBCT unit subsequent to placement of radiopaque fiduciary markers over the buccal and lingual cortical plates. Electronic linear measurements of bone height were obtained using the Sirona software. Physical measurements were obtained with digital calipers at the same location. This distance was compared on all six specimens bilaterally to determine accuracy of the image measurements.
The findings showed no statistically significant difference between the imaging and physical measurements (P > 0.05) as determined by a paired sample t-test. The intraclass correlation was used to measure the intrarater reliability of repeated measures and there was no statistically significant difference between measurements performed at the same location (P > 0.05).
The Galileos CBCT image-based linear measurement between anatomical structures within the mandible in the presence of soft tissues is sufficiently accurate for clinical use.
tomography, cone-beam computed; cadaver; head
The aim of this study was to investigate the accuracy and reliability of linear measurements of edentulous ridges recorded from 16-row multidetector CT (MDCT) images and cone beam CT (CBCT) images acquired using a flat panel detector (FPD) with a large field of view (FOV), both independently and in comparison with each other.
Edentulous areas of human dry skulls were marked with gutta-percha markers to standardize the plane of the transverse cross-sections and path of measurements. The skulls were imaged using a 16-row MDCT scanner and a CBCT device with a large FOV and a FPD. Ridge dimensions were recorded from reformatted sections by two observers and compared with measurements recorded directly from the bone. The measurement errors and intra and interexaminer reliability were calculated for each modality and compared with each other.
The overall mean of the absolute errors was 0.75 mm for MDCT and 0.49 mm for CBCT. The mean of the CBCT absolute errors was smaller than that of the MDCT absolute errors for the overall data, as well as for the site-specific data. The intraexaminer reliability score was 0.994 for MDCT and 0.995 for CBCT. The interexaminer reliability was 0.985 for MDCT and 0.958 for CBCT.
Both MDCT and CBCT were associated with a clinically and statistically significant measurement error. CBCT measurements were significantly more accurate than those of MDCT. The measurements recorded from both modalities had a high inter and intraexaminer reliability. Accuracy of measurements was found to be more operator dependent with CBCT than with MDCT.
cone beam CT; spiral CT; dental implants; reproducibility of results
The aim of this study was to investigate the possibility of reducing patient X-ray dose in the course of implant site evaluation.
Retrospective practice-based study using a Morita F170 Accuitomo cone beam CT (CBCT) scanner with variable exposure parameters and operating a small cylindrical field of view of 4 cm diameter and 4 cm height. 6 experienced dental surgeons scored the image quality of dental scans on a 5-point scale for adequacy in providing the required information in 2 categories: bone height from alveolar crest to the relevant anatomical structure and bone width.
Lower-dose protocols only marginally affected the preference of the reviewers of the resulting images.
There is potential to reduce patient dose very significantly in CBCT examinations for implant site evaluation.
cone beam computed tomography; radiography; radiation dose; dental implant
The study aimed to assess the accuracy and reproducibility of occlusal caries depth measurements obtained from different imaging modalities. The study comprised 21 human mandibular molar teeth with occlusal caries. Teeth were imaged using film, CCD, two different cone-beam computerized tomography (CBCT) units and a microcomputer tomography (micro-CT). Thereafter, each tooth was serially sectioned, and the section with the deepest carious lesion was scanned using a high-resolution scanner. Each image set was separately viewed by three oral radiologists. Images were viewed randomly, and each set was viewed twice. Lesion depth was measured on film images using a digital caliper, on CCD and CBCT images using built-in measurement tools, on micro-CT images using the Mimics software program, and on histological images using AxioVision Rel. 4.7. Intra- and inter-rater reliabilities were assessed according to the Bland/Altman method by calculating Intraclass Correlation Coefficients (ICCs). Mean/median values obtained with intraoral systems were lower than those obtained with 3-D and histological images for all observers and both readings. Intra-observer ICC values for all observers were highest for histology and micro-CT. In addition, intra-observer ICC values were higher for histology and CBCT than for histology and intra-oral methods. Inter-observer ICC values for first and second readings were high for all observers. No differences in repeatability were found between Accuitomo and Iluma CBCT images or between intra-oral film and CCD images. Micro-CT was found to be the best imaging method for the ex vivo measurement of occlusal caries depth. In addition, both CBCT units performed similarly and better than intra-oral modalities.
Occlusal caries; Depth measurement; CBCT; Micro-CT; Radiography
Mammography is the only technique currently used for detecting microcalcification (MC) clusters, an early indicator of breast cancer. However, mammographic images superimpose a three-dimensional compressed breast image onto two-dimensional projection views, resulting in overlapped anatomical breast structures that may obscure the detection and visualization of MCs. One possible solution to this problem is the use of cone beam computed tomography (CBCT) with a flat-panel (FP) digital detector. Although feasibility studies of CBCT techniques for breast imaging have yielded promising results, they have not shown how radiation dose and x-ray tube voltage affect the accuracy with which MCs are detected by CBCT experimentally. We therefore conducted a phantom study using FP-based CBCT system with various mean glandular doses and kVp values. An experimental CBCT scanner was constructed with a data-acquisition rate of 7.5 frames/s. 10.5- and 14.5cm-diameter breast phantoms made of gelatin were used to simulate uncompressed breasts consisting of 100% glandular tissue. Eight different MC sizes of calcium carbonate grains, ranging from 180–200 µm to 355–425 µm, were used to simulate MCs. MCs of the same size were arranged to form a 5×5 MC cluster and embedded in the breast phantoms. These MC clusters were positioned at 2.8 cm away from the center of the breast phantoms. The phantoms were imaged at 60, 80, and 100 kVp. With a single scan (360 degrees), 300 projection images were acquired with 0.5×, 1×, and 2× mean glandular dose limit for 10.5-cm phantom and with 1×, 2×, and 4× for 14.5-cm phantom. Feldkamp algorithm with a pure ramp filter was used for image reconstruction. The normalized noise level was calculated for each x-ray tube voltage and dose level. The image quality of CBCT images was evaluated by counting the number of visible MCs for each MC cluster for various conditions. The average percentage of the visible MCs were computed and plotted as a function of the MGD, the kVp, and the average MC size. The results show that the MC visibility increased with the MGD significantly but decreased with the breast size. The results also show the x-ray tube voltage affects the detection of MCs under different circumstances. With a 50% threshold, the minimum detectable MC sizes for the 10.5-cm phantom were 348 (±2), 288 (±7), 257 (±2) µm at 3, 6, and 12 mGy respectively. Those for the 14.5-cm phantom were 355 (±1), 307 (±7), 275 (±5) µm at 6, 12, and 24 mGy, respectively. With a 75% threshold, the minimum detectable MC sizes for the 10.5-cm phantom were 367 (±1), 316 (±7), 265 (±3) µm at 3, 6, and 12 mGy, respectively. Those for the 14.5-cm phantom were 377 (±3), 334 (±5), 300 (±2) µm at 6, 12, and 24 mGy, respectively.
cone-beam computed tomography; breast imaging; flat-panel detector; microcalcifications; mean glandular dose
The aim was to investigate the possibility of evaluating the modulation transfer function (MTF) of cone beam CT (CBCT) for dental use using the oversampling method.
The CBCT apparatus (3D Accuitomo) with an image intensifier was used with a 100 μm tungsten wire placed inside the scanner at a slight angle to the plane perpendicular to the plane of interest and scanned. 200 contiguous reconstructed images were used to obtain the oversampling line-spread function (LSF). The MTF curve was obtained by computing the Fourier transformation from the oversampled LSF. Line pair tests were also performed using Catphan®.
The oversampling method provided smooth and reproducible MTF curves. The MTF curves revealed that the spatial resolution in the z-axis direction was significantly higher than that in the axial direction. This result was also confirmed by the line pair test.
MTF analysis was performed successfully using the oversampling method. In addition, this study clarified that the 3D Accuitomo had high spatial resolution, especially in the z-axis direction.
dental; cone beam; computed tomography; modulation transfer function
The aim of this study was to assess the accuracy and reliability of cone beam CT (CBCT) images compared with multidetector CT (MDCT) images for the detection of surface osseous changes in temporomandibular joints (TMJs).
Naked-eye inspection of 110 sites in 10 TMJs from 5 dry human skulls provided the gold standard. Two radiologists interpreted the images. Sensitivity, specificity and kappa statistics were used for analysis.
The sensitivities of both modalities were low and comparable whereas the specificities were high and comparable. Intraobserver reliabilities for CBCT (p=0.0005) and for MDCT (p=0.0001) showed significant agreement. Interobserver reliability was higher for CBCT than for MDCT.
CBCT and MDCT accuracy was comparable in detecting surface osseous changes with comparable intraobserver reliabilities. However, since CBCT requires less radiation exposure, it should be encouraged for imaging TMJ with suspected surface osseous changes.
temporomandibular joint; multidetector computed tomography; cone beam computed tomography; osseous changes
Superimposition of serial Cone Beam Computed Tomography (CBCT) scans has become a valuable tool for three dimensional (3D) assessment of treatment effects and stability. Voxel based image registration is a newly developed semi-automated technique for superimposition and comparison of two CBCT scans. The accuracy and reproducibility of CBCT superimposition on the anterior cranial base or the zygomatic arches using voxel based image registration was tested in this study. 16 pairs of 3D CBCT models were constructed from pre and post treatment CBCT scans of 16 adult dysgnathic patients. Each pair was registered on the anterior cranial base three times and on the left zygomatic arch twice. Following each superimposition, the mean absolute distances between the 2 models were calculated at 4 regions: anterior cranial base, forehead, left and right zygomatic arches. The mean distances between the models ranged from 0.2 to 0.37 mm (SD 0.08–0.16) for the anterior cranial base registration and from 0.2 to 0.45 mm (SD 0.09–0.27) for the zygomatic arch registration. The mean differences between the two registration zones ranged between 0.12 to 0.19 mm at the 4 regions. Voxel based image registration on both zones could be considered as an accurate and a reproducible method for CBCT superimposition. The left zygomatic arch could be used as a stable structure for the superimposition of smaller field of view CBCT scans where the anterior cranial base is not visible.
Cross- sectional tomograms have been used for optimal pre-operative planning of dental implant placement. The aim of the present study was to assess the accuracy of Cone Beam Computed Tomography (CBCT) measurements of specific distances around the mandibular canal by comparing them to those obtained from Multi-Slice Computed Tomography (MSCT) images.
Materials and Methods:
Ten hemi-mandible specimens were examined using CBCT and MSCT. Before imaging, wires were placed at 7 locations between the anterior margin of the third molar and the anterior margin of the second premolar as reference points. Following distances were measured by two observers on each cross-sectional CBCT and MSCT image: Mandibular Width (W), Length (L), Upper Distance (UD), Lower Distance (LD), Buccal Distance (BD), and Lingual Distance (LID). The obtained data were evaluated using SPSS software, applying paired t-test and intra-class correlation coefficient (ICC).
There was a significant difference between the values obtained by MSCT and CBCT measurement for all areas such as H, W, UD, LD, BD, and LID, (P < 0.001), with a difference less than 1 mm. The ICC for all distances by both techniques, measured by a single observer with a one week interval and between 2 observers was 99% and 98%, respectively. Comparing the obtained data of both techniques indicates that the difference between two techniques is 2.17% relative to MSCT.
The results of this study showed that there is significant difference between measurements obtained by CBCT and MSCT. However, the difference is not clinically significant.
Cone beam computed tomography linear measurements; multi-slice computed tomography
To evaluate digital tomosynthesis (DTS) technology for daily positioning of patients receiving accelerated partial breast irradiation (APBI) and to compare the positioning accuracy of DTS to 3D cone-beam computed tomography (CBCT).
Methods and Materials
Ten patients who underwent APBI were scanned daily with on-board CBCT. A subset of the CBCT projections was used to reconstruct a stack of DTS image slices. To optimize soft-tissue visibility, the DTS images were reconstructed in oblique directions so that the tumor bed, breast tissue, ribs, and lungs were well separated. Coronal and sagittal DTS images were also reconstructed. Translational shifts of DTS images were obtained on different days from the same patients and were compared to the translational shifts of corresponding CBCT images. Seventy-seven CBCT scans and 291 DTS scans were obtained from 9 evaluable patients.
Tumor beds were best visible in the oblique DTS scans. One-dimensional positioning differences between DTS and CBCT images were 0.8 to 1.7 mm for the six patients with clips present and 1.2 to 2.0 mm for the three patients without clips. Due to the limited DTS scan angle, the DTS registration accuracy along the off-plane direction is lower than the accuracy along the in-plane directions.
For patients receiving APBI, DTS localization offers comparable accuracy to CBCT localization for daily patient positioning while reducing mechanical constraints and imaging dose.
on-board imaging; digital tomosynthesis; cone-beam CT; patient localization; breast cancer; limited angle cone-beam CT
The aim of this study was to compare cephalometric measurements obtained from conventional cephalograms with total and half-skull synthesized cone beam CT (CBCT) cephalograms.
Cephalometric analyses of 30 clinically symmetric patients were conducted by a calibrated examiner on conventional and CBCT-synthesized cephalograms (total, right and left). Reproducibility was investigated using the intraclass correlation coefficient (ICC). The Bland–Altman analysis was used to assess the agreement of the measurements from each factor obtained by conventional, total, right and left CBCT-synthesized cephalograms.
The ICC was above 0.9 for most of the 40 cephalometric factors analysed, revealing similar levels of reproducibility. When the measurements obtained from conventional and CBCT-synthesized cephalograms were compared, the Bland–Altman analysis showed a strong agreement between them.
Half-skull CBCT-synthesized cephalograms offer the same diagnostic performance and equivalent reproducibility in terms of cephalometric analysis as observed in conventional and total CBCT-synthesized cephalograms.
cephalometry; cone beam computed tomography
The purpose of this study was to evaluate the effectiveness of the use of Reference Ear Plug (REP) during cone-beam computed tomography (CBCT) scan for the generation of lateral cephalograms from CBCT scan data.
Two CBCT scans were obtained from 33 adults. One CBCT scan was acquired using conventional methods, and the other scan was acquired with the use of REP. Virtual lateral cephalograms created from each CBCT image were traced and compared with tracings of the real cephalograms obtained from the same subject.
CBCT scan with REP resulted in a smaller discrepancy between real and virtual cephalograms. In comparing the real and virtual cephalograms, no measurements significantly differed from real cephalogram values in case of CBCT scan with REP, whereas many measurements significantly differed in the case of CBCT scan without REP.
Measurements from CBCT-generated cephalograms are more similar to those from real cephalograms when REP are used during CBCT scan. Thus, the use of REP is suggested during CBCT scan to generate accurate virtual cephalograms from CBCT scan data.
Cone-beam computed tomography; Virtual cephalogram; Accuracy; Reference Ear Plug
The main purpose of this study was to investigate the influence of exposure parameters on jawbone model accuracy when using cone beam CT (CBCT) and multislice CT (MSCT).
A lower and an upper edentulous human cadaver jaw were scanned using micro-CT (Skyscan 1173 high energy spiral scan micro-CT; Skyscan NV, Kontich, Belgium) at 35 μm to serve as true reference. The in vitro samples were exposed using six CBCT units and one MSCT system. CBCT exposure protocols were chosen according to clinically available settings. The variables were kilovoltage, milliamperage, voxel size and/or scan time. Image segmentation was based on local thresholds using profile lines. The resulting jawbone segmentations were registered with the reference and image processing was done to internally fill the segmentations. A point-based distance calculation was performed between the three-dimensional objects and reference scans and deviation percentages were calculated for 2 mm, 1 mm and 0.5 mm intervals.
All points of the MSCT surface models lay within a 1 mm deviation range and 98.5% within 0.5 mm compared with micro-CT. For the different CBCT systems, accuracy came close to MSCT with mean percentages of 98.9% within 1 mm deviation and 92.8% within 0.5 mm. A difference of approximately 1% between lower and upper jaws could be perceived. For the specific CBCT exposure protocols, only scan time and voxel size revealed certain significant differences.
Jawbone model accuracy using CBCT was comparable with MSCT. The surface models of the upper jaws deviated slightly more than those for lower jaws. CBCT exposure settings had a limited influence on accuracy with scan time and voxel size as the main factors.
cone beam computed tomography; surface model accuracy; segmentation; exposure protocol
To investigate the accuracy and reliability of implant site measurements, recorded from low-dose cone beam computed tomography (CBCT) images.
CBCT reformatted images of five skulls were obtained using 40, 20 and 7 s exposure protocols. From these protocols, edentulous ridge dimensions were recorded by two observers and compared with measurements recorded directly from the bone. The measurement errors and intra- and inter-examiner reliability were calculated for each exposure protocol and compared with each other.
The mean absolute errors from the 40, 20 and 7 s protocols were 0.50, 0.46, and 0.51 mm, respectively. The intra-examiner reliability scores were 0.996, 0.995 and 0.998, respectively. The inter-examiner reliability scores were 0.993, 0.998 and 0.994, respectively. There was no significant difference in accuracy or reliability between the three protocols.
In imaging of dry skulls, lowering the CBCT exposure time from 40 s to 20 or 7 s does not affect the reliability or accuracy of implant site measurements.
CBCT; Radiation dosage; Reliability; Dental implantation; Dry skull
The assessment of image quality is a crucial step in the development of a new imaging protocol. Having proposed and reported on a preliminary protocol for sialography using cone beam CT (CBCT), the purpose of this study was to further optimize this protocol by maximizing the image signal difference-to-noise ratio (SDNR) and to relate these new data to previously published dosimetric data for CBCT sialography.
An imaging phantom was constructed using samples with different concentrations of iodine and a water-immersed mandible. The CB MercuRay (Hitachi Medical Systems, Tokyo, Japan) was used to image the phantom using different peak kilovoltage (kVp) and milliamperage (mA) settings. SDNR was then calculated using the raw images based on mean pixel values (MPV) measured in selected regions of interest (ROI). Finally, a figure of merit (FOM) was calculated to examine the trade-off between image SDNR and effective radiation dose.
The SDNR demonstrated an expected increase as the kVp increased from 60 to 120. Also, images made with the higher mA setting (15) had greater SDNR. The iodine concentration also influenced the image quality such that SDNR increased with increased amounts of iodine. The calculated FOM was greatest for the technique using 80 kVp, with equivalent results for 10 mA and 15 mA.
An optimized protocol for CBCT sialography using CB MercuRay entails a 6 inch field of view with 80 kVp and 10 mA.
cone beam computed tomography; sialography
Objectives: Cone Beam Computerized Tomography (CBCT) allows the possibility of modifying some of the diagnostic tools used in orthodontics, such as cephalometry. The first step must be to study the characteristics of these devices in terms of accuracy and reliability of the most commonly used landmarks. The aims were 1- To assess intra and inter-observer reliability in the location of anatomical landmarks belonging to hard tissues of the skull in images taken with a CBCT device, 2- To determine which of those landmarks are more vs. less reliable and 3- To introduce planes of reference so as to create cephalometric analyses appropriated to the 3D reality. Study design: Fifteen patients who had a CBCT (i-CAT®) as a diagnostic register were selected. To assess the reproducibility on landmark location and the differences in the measurements of two observers at different times, 41 landmarks were defined on the three spatial axes (X,Y,Z) and located. 3.690 measurements were taken and, as each determination has 3 coordinates, 11.070 data were processed with SPSS® statistical package. To discover the reproducibility of the method on landmark location, an ANOVA was undertaken using two variation factors: time (t1, t2 and t3) and observer (Ob1 and Ob2) for each axis (X, Y and Z) and landmark. The order of the CBCT scans submitted to the observers (Ob1, Ob2) at t1, t2, and t3, were different and randomly allocated. Multiple comparisons were undertaken using the Bonferroni test. The intra- and inter-examiner ICC´s were calculated. Results: Intra- and inter-examiner reliability was high, both being ICC ≥ 0.99, with the best frequency on axis Z. Conclusions: The most reliable landmarks were: Nasion, Sella, Basion, left Porion, point A, anterior nasal spine, Pogonion, Gnathion, Menton, frontozygomatic sutures, first lower molars and upper and lower incisors. Those with less reliability were the supraorbitals, right zygion and posterior nasal spine.
Key words:Cone Beam Computed Tomography, cephalometry, landmark, orthodontics, reliability.
This study was performed to assess the compatibility of cone beam computed tomography (CBCT) synthesized cephalograms with conventional cephalograms, and to find a method for obtaining normative values for three-dimensional (3D) assessments.
Materials and Methods
The sample group consisted of 10 adults with normal occlusion and well-balanced faces. They were imaged using conventional and CBCT cephalograms. The CBCT cephalograms were synthesized from the CBCT data using OnDemand 3D software. Twenty-one angular and 12 linear measurements from each imaging modality were compared and analyzed using paired-t test.
The linear measurements between the two imaging modalities were not statistically different (p>0.05) except for the U1 to facial plane distance. The angular measurements between the two imaging modalities were not statistically different (p>0.05) with the exception of the gonial angle, ANB difference, and facial convexity.
Two-dimensional cephalometric norms could be readily used for 3D quantitative assessment, if corrected for lateral cephalogram distortion.
Cone-Beam Computed Tomography; Cephalometry
Aims: To evaluate the reliability and reproducibility of calculating the Bolton Index using cone-beam computed tomography (CBCT), and to compare this with measurements obtained using the 2D Digital Method.
Material and Methods: Traditional study models were obtained from 50 patients, which were then digitized in order to be able to measure them using the Digital Method. Likewise, CBCTs of those same patients were undertaken using the Dental Picasso Master 3D® and the images obtained were then analysed using the InVivoDental programme.
Results: By determining the regression lines for both measurement methods, as well as the difference between both of their values, the two methods are shown to be comparable, despite the fact that the measurements analysed presented statistically significant differences.
Conclusions: The three-dimensional models obtained from the CBCT are as accurate and reproducible as the digital models obtained from the plaster study casts for calculating the Bolton Index. The differences existing between both methods were clinically acceptable.
Key words:Tooth-size, digital models, bolton index, CBCT.
Background and aims
With the introduction of Cone Beam Computed Tomography (CBCT) in dentistry, a question has been raised whether the technique significantly increases the diagnostic accuracy in comparison with other techniques or not. Therefore, the present study was undertaken to assess the accuracy of CBCT modality in detecting proximal carious lesions as compared to conventional periapical radiographs.
Materials and methods
This diagnostic study was carried out on 84 human extracted molars and premolars. The teeth were mounted and divided in 28 blocks of 3 teeth. Periapical and CBCT images of teeth were obtained. Five observers scored the images for the detection of proximal carious lesions using a 2-point scale (caries, present; caries, absent). The gold standard was determined by histopathologic sections. Sensitivity, specificity, PPV, NPV and receiver operating charac-teristics (ROC) curves were traced for observers in both systems. The results were analyzed by paired t-test.
The area under the ROC curve, sensitivity, specificity, accuracy, positive and negative predictive values of CBCT images were 0.568, 0.835, 0.637, 0.714, 0.598 and 0.856, respectively. These parameters were 0.432, 0.837, 0.722, 0.77, 0.687 and 0.858 for the periapical conventional technique, respectively.
The CBCT images did not enhance detection of proximal caries in comparison with periapical images.
Cone Beam CT; periapical imaging; proximal caries