|Home | About | Journals | Submit | Contact Us | Français|
We examined the comparative performance of serum and plasma (in dipotassium EDTA) in Panbio Dengue enzyme-linked immunosorbent assays (ELISAs) for detection of non-structural protein 1 (NS1), IgM, and IgG, and a dengue/Japanese encephalitis virus (JEV) combination IgM ELISA in a prospective series of 201 patients with suspected dengue in Laos. Paired comparisons of medians from serum and plasma samples were not significantly different for Dengue IgM, and NS1 which had the highest number of discordant pairs (both 2%; P = 0.13 and P = 0.25, respectively). Comparison of qualitative final diagnostic interpretations for serum and plasma samples were not significantly different: only 1.5% (3 of 201 for Dengue/JEV IgM and Dengue IgG) and 2.0% (4 of 201; IgM and NS1) showed discordant pairs. These results demonstrate that plasma containing EDTA is suitable for use in these ELISAs.
Manufacturers of diagnostic assays make specific recommendations for the sample matrix to be used for testing because blood preservatives or anticoagulants may affect assay performance. Serum centrifuged from clotted blood is often the sample of choice because it contains no chemical additives. Instructions for many commercial enzyme-linked immunosorbent assays (ELISAs) and rapid tests for diagnosis of acute dengue and Japanese encephalitis virus (JEV) infections do not state whether plasma may be used and, if so, which anticoagulant agents, such as lithium heparin, sodium fluoride, potassium oxalate, or EDTA, are appropriate.
We have therefore examined the comparative performance of paired serum and plasma samples of four well-established and previously assessed Panbio ELISAs (Alere, Brisbane, Queensland, Australia) for detection of dengue virus non-structural protein 1 (NS1),1 IgM,2 IgG,2 and a JEV IgM3 ELISA. These kits state that the test should be performed on serum only and that the use of whole blood, plasma, or other specimen matrix has not been established.
Samples (n = 201) were prospectively collected from all patients with suspected dengue-like or JEV-like illness at Mahosot Hospital, Vientiane, Laos during August–November 2010. Ethical clearance was provided by the Ethical Review Committee of the Faculty of Medical Sciences, National University of Laos (Vientiane, Laos) and the Oxford University Tropical Ethics Research Committee (Oxford, United Kingdom).
After informed written consent was obtained, patients were admitted to the study if the responsible physician diagnosed suspected dengue, defined as an acute febrile illness with ≥ 2 of the following features: headache, retro-orbital pain, myalgia, arthralgia, rash, hemorrhagic manifestations, or leukopenia according to World Health Organization guidelines.4 Venous blood samples were collected on the day of admission (admission specimen) and on the day of discharge from hospital (convalescent specimen). Serum was prepared by centrifugation of 5 mL of whole blood that was collected into plain 5-mL polystyrene blood collection tubes sterilized with gamma irradiation (Z6744; Teklab, Sacriston, United Kingdom), allowed to clot, and then centrifuged at 2,000 × g for 10 minutes. Plasma was prepared by centrifugation, as for serum, from 5 mL whole blood collected into 5-mL blood collection tubes containing 1.75 mg of dipotassium EDTA/mL (catalog no. K6740; Teklab). The two sample types were taken from the same blood draw with the same syringe and stored in the same –80°C freezers until ELISAs were performed.
The assays assessed were the Panbio Dengue Early NS1 antigen (catalog no. E-DEN01P second generation; Alere), Panbio Dengue IgM capture (catalog no. E-DEN01M; Alere), Dengue IgG capture (catalog no. DEN02G; Alere), and Panbio Japanese Encephalitis/Dengue IgM combo (catalog no. E-JED01C; Alere) ELISAs. Serum and plasma samples were tested in duplicate on the same ELISA plate to minimize variation. All assays were performed according to the manufacturer's instructions and results (Panbio Units) and final interpretations were calculated (i.e., dengue or JEV positive, negative, or inconclusive) as per the prescribed method. Inconclusive results were considered negative.
Quantitative (Panbio units) and qualitative results (positive or negative) for paired serum and plasma samples for each ELISA were compared by using STATA version 10.0 (StataCorp LP, College Station, TX). The Wilcoxon signed-rank test for matched pairs was used to test equality of Panbio Units for each ELISA. Differences in qualitative results for final assay interpretation were assessed by using McNemar's chi-square test. The range within which one would expect 95% of the values from the paired samples to lie (i.e., limits of agreement) were calculated by using the Bland-Altman method for each ELISA.5,6 P values < 0.05 were considered significant.
Comparison of the Panbio unit values and of final interpretations (positive or negative using manufacturer's criteria) for all ELISAs (Table 1) demonstrated no significant differences, with the exception of the JEV/Dengue Combo IgM ELISA, which showed significantly different results for plasma and serum for the Panbio unit comparison (P = 0.02) but not for the final interpretation (P = 0.5). There were 1.5% (3 of 201) discordant pairs for the IgG capture ELISA and 2.0% (4 of 201) discordant pairs for the IgM capture and NS1 antigen ELISAs (Table 1). Mean differences for serum and plasma Panbio units was generally small ranging from 0.07 (JEV Combo IgM capture ELISA) to 1.0 (JEV/Dengue Combo IgM capture ELISA) (Table 1). Bland and Altman 95% limits of agreement ranged from −9.50 to 9.74 for the IgG capture ELISA to –15.67 to 17.67 for the JEV/Dengue Combo IgM capture ELISA) (Table 1 and Figure 1). Comparison of Panbio unit results for dengue-positive and dengue-negative samples as determined by using the assay interpretation criteria (Table 2 and Figure 2). demonstrated that with the exception of the JEV/Dengue Combo IgM capture ELISA (P = 0.03), there were no significant differences.
These results suggest that plasma is a suitable sample for use in Panbio dengue and JEV ELISAs. Other studies with immunoassays for C-reactive protein,7 N-terminal pro-brain natriuretic peptide,8 and cryptococcal antigen9 have also demonstrated that results from serum and plasma samples are comparable. Other factors that can potentially affect ELISA results are temperature storage conditions and repeated freeze–thaw cycles. However, these affects appear to be less pronounced for ELISA detection of antibodies and viral antigens.10–12 Further studies are required to determine the affect of other anticoagulant agents such as lithium heparin, sodium fluoride and potassium oxalate, as well as the affect of sample timing, on anticoagulant-treated samples on assay performance.
We thank the directors, doctors, and nurses of Mahosot Hospital; the Director (Dr. Rattanaphone Phetsouvanh), and staff of the Microbiology Laboratory, Mahosot Hospital; and the Minister of Health and the Director of the Curative Department, Ministry of Health, for assistance and support during this study, which was part of the Lao-Oxford-Mahosot Hospital-Wellcome Trust Research Unit.
Financial support: The study was supported by the Wellcome Trust of the United Kingdom. Alere supplied the kits used in this study.
Disclosure: None of the authors have any conflicts of interest, and Alere had no role in the analysis, writing, or decision to publish these data.
Authors' addresses: Stuart D. Blacksell, Mahidol-Oxford Tropical Medicine Research Unit, Faculty of Tropical Medicine, Mahidol University, Rajthevee, Bangkok 10400, Thailand, Centre for Clinical Vaccinology and Tropical Medicine, University of Oxford, Churchill Hospital, Oxford OX3 7LJ, United Kingdom; and Lao-Oxford-Mahosot Hospital-Wellcome Trust Research Unit, Microbiology Laboratory, Mahosot Hospital, Vientiane, Laos, E-mail: stuart/at/tropmedres.ac. Sue J. Lee, Mahidol-Oxford Tropical Medicine Research Unit, Faculty of Tropical Medicine, Mahidol University, Rajthevee, Bangkok 10400, Thailand; and Centre for Tropical Medicine, University of Oxford, Churchill Hospital, Oxford OX3 7LJ, United Kingdom, E-mail: sue/at/tropmedres.ac. Anisone Chanthongthip and Soulignasack Thongpaseuth, Lao-Oxford-Mahosot Hospital-Wellcome Trust Research Unit, Microbiology Laboratory, Mahosot Hospital, Vientiane, Laos, E-mails: anisone/at/tropmedres.ac and sak/at/tropmedres.ac. Tanja Hübscher, Travel and Tropical Medicine and General Medicine, Hohmad Clinic, Thun, Switzerland, E-mail: huebscher/at/yahoo.com. Paul N. Newton, Centre for Tropical Medicine, University of Oxford, Churchill Hospital, Oxford OX3 7LJ, United Kingdom; and Lao-Oxford-Mahosot Hospital-Wellcome Trust Research Unit, Microbiology Laboratory, Mahosot Hospital, Vientiane, Laos, E-mail: paul/at/tropmedres.ac.