Revascularization with nitinol stent implantation has better long-term results than angioplasty alone [2
]; however, the cumulative incidence of femoropopliteal stent fracture varies from 2 to 65% in several studies looking specifically for stent fractures [3
]. Stent fractures can be classified into five types. Type I fracture involves only a single strut; type II involves multiple struts that can occur at different sites; type III involves multiple strut fractures resulting in a complete transverse fracture without displacement; type IV results in a complete transverse linear fracture with stent displacement; and type V is spiral fracture. Mild stent fractures rarely lead to complications, however, multiple, migrated severe stent fractures are associated with restenosis or re-occlusion at the stented site.
Stent fractures occur as a result of a combination of several factors; physical and anatomical environment, procedural choice, and structural characteristics of the stent itself [1
]. SFA is the most common site of infrainguinal PAOD, but revascularization using an endovascular stent in the SFA is subject to some vulnerable points in this physical and anatomical environment. The superficial anatomic course of the SFA is subject to external compression or torsion. The mechanical force exerted by the surrounding thigh muscle may cause stent compression or fracture. Furthermore, atherosclerosis usually involves long segments of the SFA. Scheinert et al. [4
] reported that stent fracture incidence increases with stent length, with the fracture rate being significantly lower in segments less than 8 cm. Procedural choice such as use of overlapping stents has a role in stent fracture. Stent overlapping also makes the stent susceptible to fracture. The overlapped segment may act as fulcrum for stent fracture by focusing the stress on a specific stent site [5
]. Structural characteristics of the stent also contribute to stent fracture when implanted in a specific setting. Nitinol stents, the most commonly used self-expanding stents, have great biocompatibility, elasticity and shape-memory properties; however their super-elasticity limits the stiffness of the stented vessel creating a hinge point predisposing to stent fracture [6
Restenosis, occlusion and pseudoaneurysm are the complications associated with stent fracture in SFA. These complications may lead to recurrent symptoms, and revascularization may be required. Revascularization may be achieved by re-intervention or bypass surgery. Angioplasty with or without re-stent implantation constitutes the majority of interventions. However, at times, re-canalization of the occluded stent is impossible. In this case, organization and fibrotic thrombotic in-stent material is the main cause, and many attempts for treatment have been reported by several authors. Duterloo et al. [7
] reported subintimal double-barrel restenting of an occluded primary stented SFA, they passed a guide wire through the subintima of the occluded stent, and created a neo-lumen side-by-side with the occluded primary stent leading to native lumen. Two self-expandable stents were positioned in a double-barrel fashion to prevent elastic recoil. Otherwise, Palena et al. [8
] reported a direct stent puncture technique, which consisted in puncturing directly at the occluded point and introduction of a guide wire with a J-shaped tip. Another proximal site was punctured and the guide wire was snared through the occluded point recanalizing the occluded lumen with a satisfactory angiographic result.
Early detection of stent fracture and close follow-up might allow endovascular revascularization of stent obstruction. Although there is no universal consensus, Jaff et al. [9
] suggests a standardized stent surveillance program in which high resolution digital X-ray images are obtained, and follow-up images at 12, 36, 60 months after stent deployment are compared to the baseline image obtained following stenting. At each time point, it is recommended to obtain at least 2 views of the stent with different anatomic positions of the limb such as extended and flexed knee.
In conclusion, femoral artery stenting is a common treatment for PAOD. However, stent fracture is one of the factors limiting long-term patency. We experienced two cases of stent fractures and recognized that a stent surveillance program is important for early detection and treatment before stent occlusion.