1. Machado A, Rezai AR, Kopell BH, Gross RE, Sharan AD, Benabid AL. Deep brain stimulation for Parkinson's disease: Surgical technique and perioperative management. Movement Disorders. 2006 Jun;21:S247–258. [PubMed] 2. Benabid AL, Deuschl G, Lang AE, Lyons KE, Rezai AR. Deep brain stimulation for Parkinson's disease. Movement Disorders. 2006 Jun;21(no. Suppl 14):S168–170. [PubMed] 3. Baker KB, Tkach LA, Phillips MD, Rezai AR. Variability in RF-induced heating of a deep brain stimulation implant across MR systems. J Magn Reson Imag. 2006 Dec;24:1236–1242. [PubMed] 4. Arantes PR, Cardoso EF, Barreiros MA, Teixeira MJ, Goncalves MR, Barbosa ER, Sukwinder SS, Leite CC, Amaro E., Jr Performing functional magnetic resonance imaging in patients with Parkinson's disease treated with deep brain stimulation. Movement Disorders. 2006 Aug;21:1154–1162. [PubMed] 5. Rezai AR, Baker KB, Tkach JA, Phillips M, Hrdlicka G, Sharan AD, Nyenhuis J, Ruggieri P, Shellock FG, Henderson J. Is magnetic resonance imaging safe for patients with neurostimulation systems used for deep brain stimulation? Neurosurgery. 2005 Nov;57:1056–1062. [PubMed] 6. Rezai AR, Phillips M, Baker KB, Sharan AD, Nyenhuis J, Tkach J, Henderson J, Shellock FG. Neurostimulation system used for deep brain stimulation (DBS): MR safety issues and implications of failing to follow safety recommendations. Invest Radiol. 2004 May;39:300–303. [PubMed] 7. Henderson JM, Tkach J, Phillips M, Baker K, Shellock FG, Rezai AR. Permanent neurological deficit related to magnetic resonance imaging in a patient with implanted deep brain stimulation electrodes for Parkinson's disease: Case report. Neurosurgery. 2005 Nov;57:E1063. [PubMed]
8. International Standard, Medical Equipment–Part 2–33: Particular Requirements for the Safety of the Magnetic Resonance Equipment for Medical Diagnosis, 60601-2-33. International Electrotechnical Commission; Geneva, Switzerland: 2002.
9. Bernardi P, Cavagnaro M, Pisa S, Piuzzi E. Specific absorption rate and temperature elevation in a subject exposed in the far-field of radio-frequency sources operating in the 10–900 MHz range. IEEE Trans Biomed Eng. 2003 Mar;50(no. 3):295–304. [PubMed] 10. Nguyen UD, Brown JS, Chang IA, Krycia J, Mirotznik MS. Numerical evaluation of heating of the human head due to magnetic resonance imaging. IEEE Trans Biomed Eng. 2004 Aug;51(no. 8):1301–1309. [PubMed] 11. Collins CM, Liu W, Wang J, Gruetter R, Vaughan JT, Ugurbil K, Smith MB. Temperature and SAR calculations for a human head within volume and surface coils at 64 and 300 MHz. J Magn Reson Imag. 2004 May;19:650–656. [PubMed] 12. Nitz WR, Brinker G, Diehl D, Frese G. Specific absorption rate as a poor indicator of magnetic resonance-related implant heating. Invest Radiol. 2005 Dec;40:773–776. [PubMed]
13. Guy AW. Biophysics–energy absorption and distribution. AGARD. 1974
14. Radiofrequency electromagnetic fields: Properties, quantities and units, biophysical interaction, and measurement National Council Radiation Protection and Measurements. Vol. 67. Bethesda, MD: 1981.
15. Gabriel C, Gabriel S, Corthout E. The dielectric properties of biological tissues: I. Literature survey. Phys Med Biol. 1996;41:2231–2249. [PubMed] 16. Ho HS. Safety of metallic implants in magnetic resonance imaging. J Magn Reson Imag. 2001 Oct;14:472–477. [PubMed] 17. Chou CK, McDougall JA, Chan KW. RF heating of implanted spinal fusion stimulator during magnetic resonance imaging. IEEE Trans Biomed Eng. 1997 May;44:367–373. [PubMed] 18. Armenean C, Perrin E, Armenean M, Beuf O, Pilleul F, Saint-Jalmes H. RF-induced temperature elevation along metallic wires in clinical magnetic resonance imaging: Influence of diameter and length. Magn Reson Med. 2004 Nov;52:1200–1206. [PubMed] 19. Makris N, Angelone L, Tulloch S, Sorg S, Kaiser J, Kennedy D, Bonmassar G. MRI-based anatomical model of the human head for specific absorption rate mapping. Med Biol Eng Comput. 2008 Dec;46:1239–1251. [PMC free article] [PubMed] 20. Angelone LM, Potthast A, Segonne F, Iwaki S, Belliveau JW, Bonmassar G. Metallic electrodes and leads in simultaneous EEG-MRI: Specific absorption rate (SAR) simulation studies. Bioelectro-magnetics. 2004 May;25:285–295. [PubMed]
21. Vorst Av, Rosen A, Kotsuka Y. RF/Microwave Interaction With Biological Tissues. Hoboken, NJ: Wiley: 2006. p. 73.
23. Collins CM, Smith MB. Signal-to-noise ratio and absorbed power as functions of main magnetic field strength, and definition of “90 degrees” RF pulse for the head in the birdcage coil. Magn Reson Med. 2001 Apr;45:684–691. [PubMed] 24. Mohsin SA, Sheikh NM, Saeed U. MRI-induced heating of deep brain stimulation leads. Phys Med Biol. 2008;53:5745–5756. [PubMed]
25. Li QX, Gandhi OP. Thermal implications of the new relaxed IEEE RF safety standard for head exposures to cellular telephones at 835 and 1900 MHz. IEEE Trans Microw Theory Tech. 2006 Jan;54(no. 1):3146–3154.
26. DeMarco SC, Lazzi G, Liu W, Weiland JD, Humayun MS. Computed SAR and thermal elevation in a 0.25-mm 2-D model of the human eye and head in response to an implanted retinal stimulator—Part I: Models and methods. IEEE Trans Antennas Propag. 2003 Sep;51(no. 9):2274–2285.
27. Yee KS. Numerical solution of initial boundary value problems involving Maxwell's equations in isotropic media. IEEE Trans Antennas Propag. 14 1966 Mar;(no. 3):302–307.
28. Kunz KS, Luebbers RJ. The Finite Difference Time Domain Method for Electromagnetics. Boca Raton, FL: CRC Press; 1993.
29. Berenger JP. A perfectly macthed layer for the absorption of electromagnetic waves. J Comp Phys. 1994 Oct;114:185–200.
30. Caputa K, Okoniewski M, Stuchly MA. An algorithm for computations of the power deposition in human tissue. IEEE Antennas Propagat Mag. 1999 Aug;41:102–107.
31. Ibrahim TS, Tang L, Kangarlu A, Abraham R. Electromagnetic and modeling analyses of an implanted device at 3 and 7 Tesla. J Magn Reson Imag. 2007 Nov;26:1362–1367. [PubMed]
32. Balanis CA. Antenna Theory: Analysis and Design. 3rd. Hoboken, NJ: Wiley; 2005. p. 443.
33. Bassen H, Kainz W, Mendoza G, Kellom T. MRI-induced heating of selected thin wire metallic implants– Laboratory and computational studies– Findings and new questions raised. Minimally Invasive Therory Allied Technol. 2006;15:76–84. [PubMed] 34. Vasios CE, Angelone LM, Purdon PL, Ahveninen J, Belliveau JW, Bonmassar G. EEG/(f)MRI measurements at 7 Tesla using a new EEG cap (“InkCap”) Neuroimage. 2006 Dec;33:1082–1092. [PubMed]
35. Bonmassar G. Resistive tapered stripline (RTS) in electroencephalogram recordings during MRI. IEEE Trans Microw Theory Tech. 2004 Aug;52:1992–1998.
36. Shaw RK, Long B, Werner D, Gavrin A. The characterization of conductive textile materials intended for radio frequency applications. IEEE Antennas Propag Mag. 2007 Jun;49:28–40.
37. Larsen E, Ries F. Design and calibration of the NBS isotropic electric field monitor (EFM-5), 0.2to 1000 MHz. Nat Bureau Std Tech Note Rep. 1981;1033
38. Smith G. Analysis of miniature electric field probes with resistive transmission lines. IEEE Trans Microw Theory Tech. 1981 Nov;29(no. 11):1213–1224.
39. Reilly JP. Applied Bioelectricity: From Electrical Stimulation to Electropathology. New York: Springer Verlag; 1998. p. 412.
40. Grill W. Safety considerations for deep brain stimulation: Review and analysis. Expert Rev Med Devices. 2005;2:409–420. [PubMed]
41. Bernstein M, Ward H, McGee K, Felmlee J, Knudsen1 V, Gorny K, Huston J, III, Lee K. Changes in RF transmit gain before and after DBS lead placement in 36 consecutive patients. Proc Int Soc Mag Reson Med. 2009;17:3105.
42. Alecci M, Collins CM, Smith MB, Jezzard P. Radio frequency magneticfield mapping of a 3 Tesla birdcage coil: Experimental and theoretical dependence on sample properties. Magn Reson Med. 2001 Aug;46:379–385. [PubMed]
43. Liu W, Collins CM, Smith MB. Calculations of B1 distribution, specific energy absorption rate, and intrinsic signal-to-Noise ratio for a body-size birdcage coil loaded with different human subjects at 64 and 128 MHz. Appl Magn Reson. 2005;29:5–18.
44. van den Bosch MR, Moerland MA, van Lier AL, Bartels LW, Lagendijk JJ, van den Berg CA. New method to quantify RF induced currents inside conductive wires. Proc Int Soc Mag Reson Med. 2009;17:4792.
45. Sled JG, Pike GB. Standing-wave and RF penetration artifacts caused by elliptic geometry: An electrodynamic analysis of MRI. IEEE Trans Med Imag. 1998 Aug;17:653–662. [PubMed] 46. Baker KB, Tkach J, Hall JD, Nyenhuis JA, Shellock FG, Rezai AR. Reduction of magnetic resonance imaging-related heating in deep brain stimulation leads using a lead management device. Neurosurgery. 2005;57:392–397. [PubMed] 47. Kainz W, Christ A, Kellom T, Seidman S, Nikoloski N, Beard B, Kuster N. Dosimetric comparison of the specific anthropomorphic mannequin (SAM) to 14 anatomical head models using a novel definition for the mobile phone positioning. Phys Med Biol. 2005 Jul;50:3423–3445. [PubMed]
48. Railton CJ, Schneider JB. An analytical and numerical analysis of several locally conformal FDTD schemes. IEEE Trans Microw Theory Tech. 1999 Jan;47(1):56–66.
49. Gajsek P, Walters TJ, Hurt WD, Ziriax JM, Nelson DA, Mason PA. Empirical validation of SAR values predicted by FDTD modeling. Bioelectromagnetics. 2002 Jan;23:37–48. [PubMed] 50. Jin JM, Chen J, Chew WC, Gan H, Magin RL, Dimbylow PJ. Computation of electromagnetic fields for high-frequency magnetic resonance imaging applications. Phys Med Biol. 1996 Dec;41:2719–2738. [PubMed] 51. Gandhi OP, Chen XB. Specific absorption rates and induced current densities for an anatomy-based model of the human for exposure to time-varying magnetic fields of MRI. Magn Reson Med. 1999 Apr;41:816–823. [PubMed] 52. Rezai AR, Finelli D, Nyenhuis JA, Hrdlicka G, Tkach J, Sharan A, Rugieri P, Stypulkowski PH, Shellock FG. Neurostimulation systems for deep brain stimulation: In vitro evaluation of magnetic resonance imaging-related heating at 1.5 Tesla. J Magn Reson Imag. 2002 Nov;15:241–250. [PubMed] 53. Angelone LM, Vasios CE, Wiggins G, Purdon PL, Bonmassar G. On the effect of resistive EEG electrodes and leads during 7 T MRI: Simulation and temperature measurement studies. Magn Reson Imag. 2006 Jul;24:801–812. [PubMed] 54. Ibrahim TS, Lee R, Abduljalil AM, Baertlein BA, Robitaille PM. Dielectric resonances and B(1) field inhomogeneity in UHFMRI: Computational analysis and experimental findings. Magn Reson Imag. 2001 Oct;19:219–226. [PubMed]